Bottom drain valves can be added to most Parr reactors. These valves are particularly useful for those working with polymers or other material that must be discharged from the reactor while they are still hot and before they can solidify. These valves are also quite useful for the 1 gallon and larger vessels which are too large to conveniently lift from the heater for product recovery. Bottom valves are rarely installed on the micro and mini reactors with their small volumes and light vessel weights.
The standard bottom drain valve has a rising stem, that is flush with the inside cylinder bottom so that there is no dead space between the bottom of the vessel and the shut off point of the valve. In the fully open position the stem is retracted completely to open a clear passage for draining the vessel.
When the valve is reclosed, any material in this passage will be pushed back into the reactor by the rising stem. Valves with 3/8″ diameter clear passage are recommended for vessels with volumes from 1000 mL to 2 gallons. A 1/4″ valve is available for 600 mL and smaller vessels. High pressure and larger diameter valves are available where required.
These valves will withstand the full operating pressures and temperatures of the vessels in which they are installed in the closed position. They are available in all of the current Parr materials of construction. Users can also specify that a reactor ordered with a bottom valve shall have a tapered bottom so that it will drain easily through the valve opening.
Not all Parr reactors will accept a bottom drain valve. Since the valve extends approximately 8 inches below the bottom of the vessel, the entire vessel must be raised by this amount to accommodate the valve. This makes some models too tall for convenient bench top operation. The specification tables for each model will identify those reactors in which a bottom drain can be readily installed, and those which will not accept a bottom drain, or those which will require custom modification of the heater and support stand to accommodate a bottom valve.
|Bottom Drain Valves|
|Part No.||Opening Diameter, in||Outlet Connection,||Max. Pressure psi||Max. Temp. ºC||Seal|
|A485VB||0.20||1/4″ NPT (F)||3000||225||PTFE|
|A485VB2||0.20||1/4″ NPT (F)||3000||350||Silver|
|A465VB||0.34||3/8″ NPT (F)||2000||350||Grafoil|
|A465VB2*||0.34||3/8″ NPT (F)||2000||350||Grafoil|
|A465VB3||0.34||3/8″ NPT (F)||2000||350||Silver|
|A177VB2||0.32||3/8″ NPT (F)||5000||500||Silver|
|A296VB2||0.72||1″ NPT (F)||1900||350||Silver|
|*Set up for a Band Heater|
Needle Valves and Ball Valves
Needle valves and ball valves can also be installed as bottom outlet valves. Needle valves are generally used on the smaller reactors. While ball valves can be used for large discharge passages, they are generally limited in their operating temperature/pressure capabilities and they leave a fairly large dead space between the bottom of the vessel and the seat of the valve.
4878 Automated Liquid Sampler
Catalyst Addition Devices
Coned Pressure Fittings
Equipment for Use in Potentially Ignitable Atmospheres
External Valves and Fittings
Gas Measurement Systems
--Intermediate Supply Tanks
--High Pressure Gas Burettes
Insulated Electrical Glands
Liquids Charging Systems
--Liquid Metering Pumps
--Liquid Charging Pipettes
Safety Rupture Discs
Sample Collection Vessel
Solids Charging Systems
Spare Parts Kits
Stirrer Motor and Drives
--Gas Entrainment Impellers
Valves and Fittings
--Bottom Drain Valves
--Manual Control Valves for Compressed Gas Tanks
--Pressure Relief Valves
--Safety Check Valves
Click Here to download Options and Accessories in PDF format