Click on the heading to expand
Parr offers a number of options and accessories for use with their Oxygen Vessels.
Non-volatile samples to be tested in Parr oxygen vessels are weighed and burned in shallow capsules measuring approximately 1″ diameter and 7/16″ deep. These are available in stainless steel, fused silica/quartz, and platinum alloyed with 3% rhodium.
Stainless steel capsules (43AS) are used for most applications. When combusting samples that contain metal particles such as aluminum or magnesium, the non-metallic fused silica/quartz 43A3 Capsule or fused quartz is required. When superior corrosion resistance is needed, the Platinum Rhodium 43A5 Capsule is required.
|Combustion Capsules Ordering Guidehttps://www.parrinst.com/de/wp-admin/post.php?post=681&action=edit|
|43AS||Combustion Capsule, Stainless Steel|
|43A3||Combustion Capsule, Fused Silica/Quartz|
|43A5||Combustion Capsule, Platinum – 3% Rhodium|
|43A6||Combustion Capsule, Platinum – 3% Rhodium w/flanged rim|
|169AC||Combustion Capsule, Platinum, for the Parr 1109A and 1109X Vessel|
|208AC||Combustion Capsule, lnconel, for the 1109A and 1109X Vessel|
|217A||Heavy Combustion Capsule for 1104 Vessel|
|446A||Combustion Cup, Stainless Steel, for 1121/1122 Vessel|
Parr 45C10 Fuse Wire
is a 34 B&S gage, nickel-chromium resistance wire wound on a 10 cm card for ease in cutting a fuse to length. Platinum wire is sold by length as shown.
|Fuse Wire Ordering Guide|
|45C2||Fuse Wire, Platinum, 26 ga, priced per foot|
|45C3||Fuse Wire, Platinum 36 ga, 300 cm coil|
|45C10||Fuse Wire, Ni-alloy, 10-cm card, 500 pieces per card for oxygen vessels (3 cards/package)|
The 2901 Ignition Unit
provides the proper electric current for firing a Parr oxygen vessel from a 115 or 230 volt line. Each carries a push switch, pilot light and output binding posts for connections to 7 or 10 cm vessel fuses. Connecting cords must be ordered separately.
|Ignition Unit Ordering Guide|
|2901EB||Ignition Unit, 115v 50/60 Hz|
|2901EE||Ignition Unit, 230v 50/60 Hz|
|A468E||Ignition Cord, 3-ft., single cord with spade terminal one end, banana plug other end. Use with 1108 or 1121 Vessels. Package of two.|
Oxygen Filling Connections
These Oxygen Filling Connections
provide all of the valves, gages and couplings needed for charging Parr Oxygen Vessels from a commercial oxygen tank under manual control. They are made in three styles that are identical except for the coupling on the vessel filling hose. All units fit a standard 1A oxygen tank having a CGA 540 outlet with a right handed thread. Two pressure gages are provided. The larger gage, graduated from 0 to 55 atmospheres, shows the pressure to which the vessel has been charged, while the smaller gage shows the available pressure in the supply tank. Gas flow to the vessel is controlled by a needle valve in the gage block. There is an automatic relief valve to prevent-over-charging during the filling operation.
|Oxygen Filling Connections Ordering Guide|
|1823||Oxygen Filling Connection with flat gasket coupling for use with 1104 Vessels|
|1824||Oxygen Filling Connection with snap coupling for use with 1109A and 1109X Vessels|
|1825||Oxygen Filling Connection with O-ring coupling for use with 1108 Vessels|
|1825A||Oxygen Filling Connection with threaded connection for use with 1121/1122 Vessels|
Quartz Liner for the 1108 Oxygen Vessel
For Determining Trace Elements in Combustible Samples
Trace elements in coal, oil and other combustible samples are readily determined by AA or ICP spectroscopy following combustion in an oxygen vessel. However, trace amounts of heavy metals leached from the vessel walls and electrodes during combustion can cause problems. This can be avoided by using a platinum-lined vessel, but a platinum lining is expensive. As an alternate, Parr offers a quartz liner for the 1108 Vessel that does not provide the full protection of platinum, but is, nevertheless, an effective substitute. In this arrangement the burning sample comes in contact with only the quartz liner and platinum electrodes. Reprints of papers describing the successful use of an 1108 Vessel with quartz liner for trace analysis by AA spectroscopy are available from Parr.
The 513A Quartz Liner for the 1108 Vessel consists of a quartz cup, 61mm diameter x 86 mm deep, with a 514A Flat Quartz Cover. Holes are provided in the cover for inserting platinum electrodes which support a fused silica sample cup and a short platinum fuse wire. All of the parts needed for adding this liner are provided in the Model 1912 Quartz Liner Conversion Set.
|1912 Quartz Liner Conversion Set|
|1||4AFB||Platinum Straight Electrode|
|1||5AFB||Platinum Loop Electrode|
|2||68AC||Lock Nut, T316SS|
|1||45C3||Platinum Fuse Wire, 300 cm|
|4||43A3||Fused Silica/Quartz Combustion Capsule|
Users who do not have an 1108 Vessel should order the 1911 Oxygen Vessel Apparatus which provides a vessel complete with a quartz liner and all accessories.
|1912 Quartz Liner Ordering Guide|
|1912||Conversion set for adding platinum electrodes and a quartz liner to an 1108 Oxygen Vessel|